To save content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about saving content to .
To save content items to your Kindle, first ensure firstname.lastname@example.org
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about saving to your Kindle.
Note you can select to save to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be saved to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
The purpose of this study is the verification of intensity modulated radiation therapy (IMRT) head neck treatment planning with one-dimensional and two-dimensional (2D) dosimeters using imaging and radiation oncology core (IROC) Houston head & neck (H&N) phantom.
The image of the H&N phantom was obtained by computed tomography scan which was then transferred to Pinnacle@3 treatment planning system (TPS) for treatment planning. The contouring of the target volumes and critical organ were done manually and dose constraints were set for each organ according to IROC prescription. The plan was optimised by adoptive convolution algorithm to meet the IROC criteria and collapse cone convolution algorithm calculated the delivered doses for treatment. Varian Clinac 2110 was used to deliver the treatment plan to the phantom, the process of irradiation and measurement were repeated three times for reproducibility and reliability. The treatment plan was verified by measuring the doses from thermoluminescent dosimeters (TLDs) and GafChromic external beam therapy 2 films. The agreement between the planned and delivered doses were checked by calculating the percentage dose differences, analysing their isodose line profiles and 2D gamma maps.
The average percent dose difference of 1·8% was obtained between computed doses by TPS and measured doses from TLDs, however these differences were found to be higher for organ at risk. The film dose profile was well in agreement with the planned dose distribution with distance to agreement of 1·5 mm. The gamma analysis of the computed and recorded doses passed the criteria of 3%/3 mm with passing percentages of >96%, which shows successful authentication of delivered doses for IMRT.
IMRT pre-treatment validation can be done with IROC anthropomorphic phantoms, which is essential for the delivery of modulated radiotherapies. It was concluded that films and TLDs can be used as quality assurance tools for IMRT.
To verify dose delivery and quality assurance of volumetric-modulated arc therapy (VMAT) for head and neck (H&N) cancer.
The Imaging and Radiation Oncology Core Houston (IROC-H) H&N phantom with thermoluminescent dosimeters (TLDs) and films, were imaged with computed tomography scan and the reconstructed image was transferred to pinnacle treatment planning system (TPS). On TPS, the planning target volume (PTV), secondary target volume (STV) and organ at risk (OAR) were delineated manually and a treatment plan was made. The dose constraints were determined for the concerned organs according to IROC-H prescription. The treatment plan was optimised using adoptive convolution algorithm to improve dose homogeneity and conformity. The dose calculation was performed using C.C Convolution algorithm and a Varian True Beam linear accelerator was used to deliver the treatment plan to the H&N phantom. The delivered radiation dose to the phantom was measured through TLDs and GafChromic external beam radiotherapy 2 (EBT2) films. The dosimetric performance of the VMAT delivery was studied by analysing percent dose difference, isodose line profile and gamma analysis of the TPS-computed dose and linac-delivered doses.
The percent dose difference of 3.8% was observed between the planned and measured doses of TLDs and a 1.5-mm distance to agreement (DTA) was observed by comparing isodose line profiles. Passed the gamma criteria of 3%/3 mm was with good percentages.
The dosimetric performance of VMAT delivery for a challenging H&N radiotherapy can be verified using TLDs and films embedded in an anthropomorphic H&N phantom.