Book chapters will be unavailable on Saturday 24th August between 8am-12pm BST. This is for essential maintenance which will provide improved performance going forwards. Please accept our apologies for any inconvenience caused.
To send content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about sending content to .
To send content items to your Kindle, first ensure firstname.lastname@example.org
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about sending to your Kindle.
Note you can select to send to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be sent to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
This study reported the justification and selection of acceptable γ criteria with respect to low (6 MV) and high (15 MV) photon beams for intensity-modulated radiation therapy quality assurance (IMRT QA) using the Gafchromic external beam therapy 3 (EBT3) film.
Materials and methods
Five-field step-and-shoot IMRT was used to treat 16 brain IMRT patients using the dual-energy DHX-S linear accelerator (Varian Medical System, Palo Alto, CA, USA). Dose comparisons between computed values of the treatment planning system (TPS) and Gafchromic EBT3 film were evaluated based on γ analysis using the Film QA Pro software. The dose distribution was analysed with gamma area histograms (GAHs) generated using different γ criteria (3%/2 mm, 3%/3 mm and 5%/3 mm) for the 6 and 15 MV photon beams, to optimise the best distance-to-agreement (DTA) criteria with respect to the beam energy.
From the comparison between the dose distributions acquired from the TPS and EBT3 film, a DTA criterion of 3%/2 mm showed less dose differences (DDs) with passing rates up to 93% for the 6 MV photon beams, while for the 15 MV a relaxed DTA criterion of 5%/3 mm was consistent with the DD acceptability criteria with a 95% passing rate.
Our results suggested that high-energy photon beams required relaxed DTA criteria for the brain IMRT QA, while low-energy photon beams showed better results even with tight DTA criteria.
To verify dose delivery and quality assurance of volumetric-modulated arc therapy (VMAT) for head and neck (H&N) cancer.
The Imaging and Radiation Oncology Core Houston (IROC-H) H&N phantom with thermoluminescent dosimeters (TLDs) and films, were imaged with computed tomography scan and the reconstructed image was transferred to pinnacle treatment planning system (TPS). On TPS, the planning target volume (PTV), secondary target volume (STV) and organ at risk (OAR) were delineated manually and a treatment plan was made. The dose constraints were determined for the concerned organs according to IROC-H prescription. The treatment plan was optimised using adoptive convolution algorithm to improve dose homogeneity and conformity. The dose calculation was performed using C.C Convolution algorithm and a Varian True Beam linear accelerator was used to deliver the treatment plan to the H&N phantom. The delivered radiation dose to the phantom was measured through TLDs and GafChromic external beam radiotherapy 2 (EBT2) films. The dosimetric performance of the VMAT delivery was studied by analysing percent dose difference, isodose line profile and gamma analysis of the TPS-computed dose and linac-delivered doses.
The percent dose difference of 3.8% was observed between the planned and measured doses of TLDs and a 1.5-mm distance to agreement (DTA) was observed by comparing isodose line profiles. Passed the gamma criteria of 3%/3 mm was with good percentages.
The dosimetric performance of VMAT delivery for a challenging H&N radiotherapy can be verified using TLDs and films embedded in an anthropomorphic H&N phantom.
This work reports our study to commission a radiochromic film dosimetry system using the timely EBT3 film. We carried out dosimetric evaluations on different characteristics of photon beams (e.g., flatness, symmetry and penumbra) in radiation dose delivery.
Materials and Methods
A Varian linear accelerator producing 6 and 15 MV photon beams with 120 multi-leaf collimator was used in this study. PTW ionisation chamber was used to measure the beam characteristics such as symmetry, flatness and penumbra and these measurements were used to commission the radiochormic EBT3 film dosimetry system. The results of irradiated films were analysed using the radiochromic film QA Pro software 2016.
The measured film doses were analysed at two different colour channels (green and red) using two scanning geometries (i.e., upper or lower side of film facing the scanner light source) at two dose levels (10 and 40 Gy). The difference between the ionisation chamber and film results was found insignificant and within the acceptable range as per the World Health Organisation standard.
Results of the comparison between the ionisation chamber and film measurements show that our radiochormic EBT3 film dosimetry system is reliable and cost-effective in the output measurement of a linear accelerator. Our measurements confirm that our EBT3 film dosimetry agreed well with the ionisation chamber, and can be used as a re-validation tool for linear accelerator quality control.
A survey of alcohol dehydrogenase (Adh) in Culex tritaeniorhynchus Giles has uncovered five electrophoretic variants among laboratory strains. Detailed genetic analysis revealed that these variants are inherited as codominant alleles at a single locus. The Adh locus has been assigned to linkage group III between s and cl. A survey of a natural population from Lahore, Pakistan, revealed the presence of four of the five variants and suggests the possibility of using this enzyme polymorphism to monitor a release programme. Substrate specificity tests suggest that this enzyme can utilize almost any aliphatic primary alcohol as a substrate.
Email your librarian or administrator to recommend adding this to your organisation's collection.