To save content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about saving content to .
To save content items to your Kindle, first ensure firstname.lastname@example.org
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about saving to your Kindle.
Note you can select to save to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be saved to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
We compared sepsis “time zero” and Centers for Medicare and Medicaid Services (CMS) SEP-1 pass rates among 3 abstractors in 3 hospitals. Abstractors agreed on time zero in 29 of 80 (36%) cases. Perceived pass rates ranged from 9 of 80 cases (11%) to 19 of 80 cases (23%). Variability in time zero and perceived pass rates limits the utility of SEP-1 for measuring quality.
Endoscope-associated infections are reported despite following proper reprocessing methods. Microbiological testing can confirm the adequacy of endoscope reprocessing. Multiple controversies related to the method and interpretation of microbiological testing cultures have arisen that make their routine performance a complex target.
We conducted a pilot study using disposable bronchoscopes (DBs) to simulate different reprocessing times and soaking times and to compare high-level disinfection versus ethylene oxide sterilization. We also reviewed the time to reprocessing and duration of the procedures.
Bronchoscopes were chosen because an alternative disposable scope is commercially available and because bronchoscopes are more prone to delays in processing. Disposable bronchoscopes were contaminated using a liquid bacterial suspension and were then incubated for 1–4 hours. Standard processing and high-level disinfection were performed on 36 endoscopes. Ethylene oxide sterilization was performed on 21 endoscopes. Endoscope cultures were performed using the standard “brush, flush, brush” technique.
After brushing was performed, a final water-flush culture procedure was the most effective method of detecting bacterial persistence on the disposable scopes. Klebsiella pneumoniae was the most commonly recovered organism after reprocessing. Ethylene oxide sterilization did not result in total elimination of viable bacteria.
Routine endoscopy cultures may be required to assess the adequacy of endoscopic processing.
Infect Control Hosp Epidemiol 2017;38:136–142
Email your librarian or administrator to recommend adding this to your organisation's collection.