To save content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about saving content to .
To save content items to your Kindle, first ensure email@example.com
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about saving to your Kindle.
Note you can select to save to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be saved to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
To investigate the feasibility of using an ultraviolet light-emitting diode (UV LED) robot for the terminal decontamination of coronavirus disease 2019 (COVID-19) patient rooms.
We assessed the presence of viral RNA in samples from environmental surfaces before and after UV LED irradiation in COVID-19 patient rooms after patient discharge.
We analyzed 216 environmental samples from 17 rooms: 2 from airborne infection isolation rooms (AIIRs) in the intensive care unit (ICU) and 15 from isolation rooms in the community treatment center (CTC). Severe acute respiratory syndrome coronavirus 2 (SARS-CoV-2) RNA was detected in 40 (18.5%) of 216 samples after patient discharge: 12 (33.3%) of 36 samples from AIIRs in the ICU, and 28 (15.6%) of 180 samples from isolation rooms in the CTC. In 1 AIIR, all samples were PCR negative after UV LED irradiation. In the CTC rooms, 14 (8.6%) of the 163 samples were PCR positive after UV LED irradiation. However, viable virus was not recovered from the culture of any of the PCR-positive samples.
Although no viable virus was recovered, SARS-CoV-2 RNA was detected on various environmental surfaces. The use of a UV LED disinfection robot was effective in spacious areas such as an ICU, but its effects varied in small spaces like CTC rooms. These findings suggest that the UV LED robot may need enough space to disinfect rooms without recontamination by machine wheels or insufficient disinfection by shadowing.
Email your librarian or administrator to recommend adding this to your organisation's collection.