To save content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about saving content to .
To save content items to your Kindle, first ensure email@example.com
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about saving to your Kindle.
Note you can select to save to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be saved to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
A provisional setup for X-ray microprobe experiments at 35 keV is described. It is based on compound refractive lenses (CRLs) for nanofocusing and a Vortex silicon drift detector with 2 mm sensor thickness for increased sensitivity at high energies. The Microprobe experiment (PETRA III) generally uses Kirkpatrick-Baez mirrors for submicrometer focusing in the energy range of 5–21 keV. However, various types of scanning X-ray microscopy experiments require higher excitation energies. The CRL optics were characterized by X-ray ptychography and X-ray fluorescence (XRF) knife edge scans on a siemens star pattern and showed beam sizes down to 110 nm. The performance of the new setup for microscopic X-ray diffraction (XRD)–XRF scanning X-ray microscopy measurements at 35 keV is demonstrated on a cross-section of a painting fragment.