To save content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about saving content to .
To save content items to your Kindle, first ensure firstname.lastname@example.org
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about saving to your Kindle.
Note you can select to save to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be saved to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
Transcatheter ventricular septal defect closure remains a complex procedure with potential complications like complete heart block and aortic regurgitation. The ideal device design for such intervention is still evolving.
To assess the safety, efficacy, and short-term outcome of ventricular septal defect closure using LifeTechTM multifunctional (KONAR-MFTM) VSD Occluder.
Patients and methods:
In a multicenre study, 44 patients with haemodynamically significant, restrictive ventricular septal defects underwent closure with the KONAR-MFTM device from April, 2019 to March, 2020. Clinical, echocardiographic, and angiographic data were collected and reviewed. Patients were followed up at 1, 3, 6, and 12 months.
The median age and weight were 8 (1.7–36) years and 20 (11–79) kg. Of 44 patients, 8 (18%) had a high muscular and 36 (82%) had a perimembranous defect, of which 6 had mild prolapse of the right coronary cusp. The median ventricular septal defect size was 8.8 (3.9–13.4) mm. A retrograde approach was adopted in 39 (88.6%) patients. Nine patients (20.5%) had a small residual leak and there was a slight increase in aortic regurgitation in one patient. One device, which embolised to pulmonary artery was retrieved, and the defect was closed with a larger device. At a median follow-up of 13 (5–18) months, the residual leak persisted in 1 (2.3%) patient. Mild aortic regurgitation in one patient remained unchanged. There were no major complications.
Percutaneous closure of ventricular septal defect using KONAR-MFTM device is safe and effective in short and midterm follow-up including selected patients with perimembranous defect and mild prolapse of the right coronary cusp.
Email your librarian or administrator to recommend adding this to your organisation's collection.