To save content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about saving content to .
To save content items to your Kindle, first ensure firstname.lastname@example.org
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about saving to your Kindle.
Note you can select to save to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be saved to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
This study mainly focused on the key technologies, the photon dose calculation based on the Monte Carlo Finite-Size Pencil Beam (MCFSPB) model in the Accurate Radiotherapy System (ARTS). In the MCFSPB model, the acquisition of pencil beam kernel is one of the most important technologies. In this study, by analyzing the demerits of the clinical pencil beam dose calculation methods, a new pencil beam kernel model was developed based on the Monte Carlo (MC) simulation and the technology of medical accelerator energy spectrum reconstruction. which greatly improved the accuracy of calculated result. According to the axial symmetry principle, only part of simulation results was used for the data of pencil beam kernel, which greatly reduced the data quantity of the pencil beam and reduced calculated time. Based on the above studies, the MCFSPB method was designed and implemented by the Visual C++ development tool. With several tests including the comparisons among the American Association of Physicists in Medicine (AAPM) No. 55 Report sample and the ion chamber measurement of lung-simulating inhomogeneous phantom in clinical treatment plan, the results showed that the maximum error of most calculated point was less than 0.5% in the homogeneous phantom and less than 3% in the heterogeneous phantom. This method met the clinical criteria, and would be expected to be used as a fast and accurate dose engine for clinic TPS.
Email your librarian or administrator to recommend adding this to your organisation's collection.