To save content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about saving content to .
To save content items to your Kindle, first ensure firstname.lastname@example.org
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about saving to your Kindle.
Note you can select to save to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be saved to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
The purpose of this investigation was to expand upon the limited existing research examining the test–retest reliability, cross-sectional validity and longitudinal validity of a sample of bioelectrical impedance analysis (BIA) devices as compared with a laboratory four-compartment (4C) model. Seventy-three healthy participants aged 19–50 years were assessed by each of fifteen BIA devices, with resulting body fat percentage estimates compared with a 4C model utilising air displacement plethysmography, dual-energy X-ray absorptiometry and bioimpedance spectroscopy. A subset of thirty-seven participants returned for a second visit 12–16 weeks later and were included in an analysis of longitudinal validity. The sample of devices included fourteen consumer-grade and one research-grade model in a variety of configurations: hand-to-hand, foot-to-foot and bilateral hand-to-foot (octapolar). BIA devices demonstrated high reliability, with precision error ranging from 0·0 to 0·49 %. Cross-sectional validity varied, with constant error relative to the 4C model ranging from −3·5 (sd 4·1) % to 11·7 (sd 4·7) %, standard error of the estimate values of 3·1–7·5 % and Lin’s concordance correlation coefficients (CCC) of 0·48–0·94. For longitudinal validity, constant error ranged from −0·4 (sd 2·1) % to 1·3 (sd 2·7) %, with standard error of the estimate values of 1·7–2·6 % and Lin’s CCC of 0·37–0·78. While performance varied widely across the sample investigated, select models of BIA devices (particularly octapolar and select foot-to-foot devices) may hold potential utility for the tracking of body composition over time, particularly in contexts in which the purchase or use of a research-grade device is infeasible.
Email your librarian or administrator to recommend adding this to your organisation's collection.