To save content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about saving content to .
To save content items to your Kindle, first ensure firstname.lastname@example.org
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about saving to your Kindle.
Note you can select to save to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be saved to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
To determine the impact of three needleless intravenous systems on needlestick injury rates.
Randomized controlled trial.
1,000-bed tertiary-care Midwestern hospital.
Nursing personnel from general medical, general surgical, and intensive-care units.
From June 1992 through March 1994, a metal blunt cannula (MBC), two-way valve (2-way), and plastic blunt cannula (PBC) were introduced into three study areas, and needlestick injury rates were compared to three control areas using traditional needled devices.
24 and 29 needlestick injuries were reported in study and control areas. Intravenous-therapy-related injuries comprised 45.8% and 57.1% of injuries in each area. Thirty-seven percent and 20.7% of study and control area needlestick injuries were considered to pose a high risk of bloodborne infection. The 2-way group had similar rates of total and intravenous-related needlestick injuries compared to control groups. The PBC group had lower rates of total and intravenous-related needlestick injuries per 1,000 patient-days (rate ratios [RR], 0.32 and 0.24; 95°% confidence intervals [CI95], 0.12-0.81 and 0.09-0.61; P=.02 and P=.003, respectively) and per 1,000 productive hours worked (RR, 0.11 and 0.08; CI95, 0.01-0.92 and 0.010.69; P=.03 and P=.005, respectively) compared to controls.
Needlestick injuries continued in study areas despite the introduction of needleless devices, and risks of bloodborne pathogen transmission were similar to control areas. The PBC device group noted lower rates of needlestick injuries compared to controls, but there were problems with product acceptance, correct product use, and continued traditional device use in study areas. Low needlestick injury rates make interpretations difficult. Further studies of safety devices are needed and should attempt greater control of worker behavior to aid interpretation.
Email your librarian or administrator to recommend adding this to your organisation's collection.