To save content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about saving content to .
To save content items to your Kindle, first ensure firstname.lastname@example.org
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about saving to your Kindle.
Note you can select to save to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be saved to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
We have developed a system using ‘forescatter detectors’ for backscattered imaging of specimen surfaces inclined at 50–80° to the incident beam (inclined-scanning) in the SEM. These detectors comprise semiconductor chips placed below the tilted specimen. Forescatter detectors provide an orientation contrast (OC) image to complement quantitative crystallographic data from electron backscatter patterns (EBSP). Specimens were imaged using two detector geometries and these images were compared to those collected with the specimen surface normal to the incident beam (normal-scanning) using conventional backscattered electron detector geometries and also to an automated technique, orientation imaging microscopy (OIM). When normal-scanning, the component of the BSE signal relating to the mean atomic number (z) of the material is an order of magnitude greater than any OC component, making OC imaging in polyphase specimens almost impossible. Images formed in inclined-scanning, using forescatter detectors, have OC and z-contrast signals of similar magnitude, allowing OC imaging in polyphase specimens.
OC imaging is purely qualitative, and by repeatedly imaging the same area using different specimen-beam geometries, we found that a single image picks out less than 60% of the total microstructural information and as many as 6 combined images are required to give the full data set. The OIM technique is limited by the EBSP resolution (1–2°) and subsequently misses a lot of microstructural information. The use of forescatter detectors is the most practical means of imaging OC in tilted specimens, but it is also a powerful tool in its own right for imaging microstructures in polyphase specimens, an essential asset for geological work.
Email your librarian or administrator to recommend adding this to your organisation's collection.