To send content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about sending content to .
To send content items to your Kindle, first ensure email@example.com
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about sending to your Kindle.
Note you can select to send to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be sent to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
To determine the frequency and factors associated with increase in microembolic signals during transcatheter closure of atrial septal defect using the Amplatzer septal occluder.
During the procedure in 16 patients, we measured microembolic signals using transcranial Doppler. Procedure time was divided into five periods: right cardiac catheterisation; left cardiac catheterisation; left cardiac angiocardiography; sizing and long sheath placement; device placement and release. We compared numbers of microembolic signals among the five periods and identified factors associated with them.
Mean size of septal occluder was 16 millimetres in diameter. Total number of microembolic signals was a median of 31.5, ranging from 3 to 113. Microembolic signals in three periods, left cardiac catheterisation; sizing, and long sheath placement; and device placement and release, were not significantly different from one another, but were significantly higher than those in the remaining two periods, right cardiac catheterisation and left cardiac angiocardiography (median was 9 in left cardiac catheterisation; 6 in sizing and long sheath placement; 6.5 in device placement and release, versus 0 in right cardiac catheterisation and 1 in left cardiac angiocardiography, p less than 0.05, respectively). Importantly, the time for device manipulation positively correlated with total number of microembolic signals (r equals 0.77, p less than 0.001), although fluoroscopic time, age, or size of septal occluder did not.
Transcatheter closure of atrial septal defect using the Amplatzer septal occluder produces microemboli, especially during device placement. To minimise the risk of systemic embolism, we must decrease the time for device manipulation.