To send content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about sending content to .
To send content items to your Kindle, first ensure email@example.com
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about sending to your Kindle.
Note you can select to send to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be sent to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
Whole-breast external beam radiotherapy results in significant reduction in the risk for breast cancer-related death, but this may be offset by an increase in deaths from other causes and toxicity to surrounding organs. Partial breast irradiation techniques are approaches that treat only the lumpectomy area rather than the whole breast. Quality assurance in the radiation therapy treatment planning process is essential to ensure accurate dose delivery to the patient. For this purpose, this article compares the results from an anthropomorphic PRESAGE® dosimeter, radiation treatment planning system and from the GAFCHROMIC® EBT2 film.
Materials and methods
A breast dosimeter was created and a three-field partial plan was generated in the Pinnacle3 treatment planning system. Dose distribution comparisons were made between Pinnacle3 treatment planning system, GAFCHROMIC® EBT2 film and PRESAGE® dosimeter. Dose–volume histograms (DVHs), gamma maps and line profiles were used to evaluate the comparison.
DVHs of gross tumour volume, clinical tumour volume and planning tumour volume for the PRESAGE® dosimeter and Pinnacle3 treatment planning system shows that both measured and calculated statistics were in agreement, with a value of 97.8% of the prescribed dose. Gamma map comparisons showed that all three distributions passed 95% at the ±3%/±3 mm criteria. Comparisons of isodose line distribution between the PRESAGE® dosimeter, EBT2 film and planning system demonstrated agreement, with an average difference of 1.5%.
This work demonstrated the feasibility of PRESAGE® to function as an anthropomorphic phantom and laid the foundation for research studies in PRESAGE®/optical-computed tomography three-dimensional dosimetry with the most complex anthropomorphic phantoms.
Email your librarian or administrator to recommend adding this to your organisation's collection.