To send content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about sending content to .
To send content items to your Kindle, first ensure email@example.com
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about sending to your Kindle.
Note you can select to send to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be sent to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
To determine the cause of acute illness on August 30, 2000, among patients at an outpatient dialysis center (center A).
We performed a cohort study of all patients receiving dialysis on August 30, 2000; reviewed dialysis procedures; and analyzed dialysis water samples using microbiologic and chemical assays.
Dialysis center (center A).
A case-patient was defined as a patient who developed chills within 5 hours after starting hemodialysis at center A on August 30, 2000.
Sixteen (36%) of 44 patients at center A met the case definition. All case-patients were hospitalized; 2 died. Besides chills, 15 (94%) of the case-patients experienced nausea; 12 (75%), vomiting; and 4 (25%), fever. Illness was more frequent on the second than the first dialysis shift (16 of 20 vs 0 of 24, P < .001); no other risk factors were identified. The center's water treatment system had received inadequate maintenance and disinfection and a sulfurous odor was noted during sampling of the water from the reverse osmosis (RO) unit. The water had elevated bacterial counts. Volatile sulfur-containing compounds (ie, methanethiol, carbon disulfide, dimethyldisulfide, and sulfur dioxide) were detected by gas chromatography and mass spectrometry in 8 of 12 water samples from the RO unit and in 0 of 28 samples from other areas (P < .001). Results of tests for heavy metals and chloramines were within normal limits.
Parenteral exposure to volatile sulfur-containing compounds, produced under anaerobic conditions in the RO unit, could have caused the outbreak. This investigation demonstrates the importance of appropriate disinfection and maintenance of water treatment systems in hemodialysis centers.
Email your librarian or administrator to recommend adding this to your organisation's collection.