To send content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about sending content to .
To send content items to your Kindle, first ensure firstname.lastname@example.org
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about sending to your Kindle.
Note you can select to send to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be sent to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
Operators are mindful of the balloon-to-aortic annulus ratio when performing balloon aortic valvuloplasty. The method of measurement of the aortic valve annulus has not been standardised.
Methods and results:
Patients who underwent aortic valvuloplasty at two paediatric centres between 2007 and 2014 were included. The valve annulus measured by echocardiography and angiography was used to calculate the balloon-to-aortic annulus ratio and measurements were compared. The primary endpoint was an increase in aortic insufficiency by ≥2 degrees. Ninety-eight patients with a median age at valvuloplasty of 2.1 months (Interquartile range (IQR): 0.2–105.5) were included. The angiographic-based annulus was 8.2 mm (IQR: 6.8–16.0), which was greater than echocardiogram-based annulus of 7.5 mm (IQR: 6.1–14.8) (p < 0.001). This corresponded to a significantly lower angiographic balloon-to-aortic annulus ratio of 0.9 (IQR: 0.9–1.0), compared to an echocardiographic ratio of 1.1 (IQR: 1.0–1.1) (p < 0.001). The degree of discrepancy in measured diameter increased with smaller valve diameters (p = 0.041) and in neonates (p = 0.044). There was significant disagreement between angiographic and echocardiographic balloon-to-aortic annulus ratio measures regarding “High” ratio of >1.2, with angiographic ratio flagging only 2/12 (16.7%) of patients flagged by echocardiographic ratio as “High” (p = 0.012). Patients who had an increase in the degree of aortic insufficiency post valvuloplasty, only 3 (5.5%) had angiographic ratio > 1.1, while 21 (38%) had echocardiographic ratio >1.1 (p < 0.001). Patients with resultant ≥ moderate insufficiency more often had an echocardiographic ratio of >1.1 than angiographic ratio of >1.1 There was no association between increase in balloon-to-aortic annulus ratio and gradient reduction.
Angiographic measurement is associated with a greater measured aortic valve annulus and the development of aortic insufficiency. Operators should use caution when relying solely on angiographic measurement when performing balloon aortic valvuloplasty.