To send content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about sending content to .
To send content items to your Kindle, first ensure firstname.lastname@example.org
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about sending to your Kindle.
Note you can select to send to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be sent to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
The aim of this study is to investigate the impact of anatomical changes in prostate cancer patients on the target coverage when using 6 MV beams-VMAT therapy and to propose strategies that allow us to evaluate the dose or correct it by normalization without having to perform a new simulation.
Methods and materials:
Ten patients of high-risk prostate cancer were chosen for the study. All test plans were delivered using the same isocenter and monitor units as the original plan and compared against the original unedited plan. The expansion and contraction of body contours due to size changes was mimicked by increasing and decreasing the body contour with depths of −2, −1·5, …, 1·5, 2 cm, in the anterior, and both lateral directions of the patient. A total of 90 plans were evaluated, 9 for each patient. Dose-volume histogram statistics were extracted from each plan and normalized to prescription dose.
Weight changes resulted in considerable dose modifications to the target and critical structures. Plans were found to be varied with 2·9% ± 0·3% per cm SSD change for VMAT treatment with a correlation index close to one. Therefore, doses variations were linear to the changes of depth. Gamma index evaluation was performed for the 10 renormalized plans. All of them passed criteria of 3%/3 mm in at least 98.2% of points. Eight of them passed criteria in 99% points. Gamma index 4%/4 mm passed 100% points in all patients for the chosen region of interest.
The dosimetry estimation presented in this study shows important data for the radiation oncology staff to justify whether a CT rescan is necessary or not when a patient experiences weight changes during treatment. Based on the results of our study, discrepancies between real dose and planned dose were >5% for 1·7 cm of difference in external contour in the anterior and both lateral directions of the patient.
Email your librarian or administrator to recommend adding this to your organisation's collection.