To send content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about sending content to .
To send content items to your Kindle, first ensure firstname.lastname@example.org
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about sending to your Kindle.
Note you can select to send to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be sent to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
In advanced radiotherapy techniques such as intensity-modulated radiation therapy (IMRT), the quality assurance (QA) process is essential. The aim of the study was to assure the treatment planning dose delivered during delivery of complex treatment plans. The QA standard is to perform patient-specific comparisons between planned doses and doses measured in a phantom.
Materials and method
The Delta 4 phantom (Scandidos, Uppsala, Sweden) has been used in this study. This device consists of diode matrices in two orthogonal planes inserted in a cylindrical acrylic phantom. Each diode is sampled per beam pulse so that the dose distribution can be evaluated on segment-by-segment, beam-by-beam, or as a composite plan from a single set of measurements. Ninety-five simple and complex radiotherapy treatment plans for different pathologies, planned using a treatment planning system (TPS) were delivered to the QA device. The planned and measured dose distributions were then compared and analysed. The gamma index was determined for different pathologies.
The evaluation was performed in terms of dose deviation, distance to agreement and gamma index passing rate. The measurements were in excellent agreement between with the calculated dose of the TPS and the QA device. Overall, good agreement was observed between measured and calculated doses in most cases with gamma values above 1 in >95% of measured points. Plan results for each test met the recommended dose goals.
The delivery of IMRT and volumetric-modulated arc therapy (VMAT) plans was verified to correspond well with calculated dose distributions for different pathologies. We found the Delta 4 device is accurate and reproducible. Although Delta4 appears to be a straightforward device for measuring dose and allows measure in real-time dosimetry QA, it is a complex device and careful quality control is required before its use.
Email your librarian or administrator to recommend adding this to your organisation's collection.