To send content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about sending content to .
To send content items to your Kindle, first ensure email@example.com
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about sending to your Kindle.
Note you can select to send to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be sent to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
The purpose of this study was to evaluate the dosimetric impact of multileaf collimator (MLC) positional errors on dynamic intensity-modulated radiotherapy (IMRT) treatments through planning simulation. Secondly the sensitivity of IMRT MatriXX device for detecting the MLC leaf positional errors was also evaluated.
Materials and methods
In this study five dynamic IMRT plans, each for brain and head–neck (HN), were retrospectively included. An in-house software was used to introduce random errors (uniform distribution between −2·0 and +2·0 mm) and systematic errors [±0·5, ±0·75, ±1·0 and ±2·0 mm (+: open MLC error and −: close MLC error)]. The error-introduced MLC files were imported into the treatment planning system and new dose distributions were calculated. Furthermore, the dose–volume histogram files of all plans were exported to in-house software for equivalent uniform dose (EUD), tumour control probability and normal tissue complication probability calculations. The error-introduced plans were also delivered on LINAC, and the planar fluences were measured by IMRT MatriXX. Further, 3%/3 mm and 2%/2 mm γ-criteria were used for analysis.
In planning simulation study, the impact of random errors was negligible and ΔEUD was <0·5±0·7%, for both brain and HN. The impact of systematic errors was substantial, and on average, the maximum change in EUD for systematic errors (close 2 mm) was −10·7±3·1% for brain and −15·5±2·6% for HN.
It can be concluded that the acceptable systematic error was 0·4 mm for brain and 0·3 mm for HN. Furthermore, IMRT MatriXX device was able to detect the MLC errors ≥2 mm in HN and >3 mm errors in brain with 2%/2 mm γ-criteria.
Email your librarian or administrator to recommend adding this to your organisation's collection.