To send content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about sending content to .
To send content items to your Kindle, first ensure email@example.com
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about sending to your Kindle.
Note you can select to send to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be sent to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
In November 2009, routine sampling of endoscopes performed to monitor the effectiveness of the endoscope-cleaning procedure at our hospital detected Pseudomonas aeruginosa. Herein we report the results of the subsequent investigation.
Design and Methods.
The investigation included environmental cultures for source investigation, molecular analysis by pulsed-field gel electrophoresis (PFGE) to reveal the identity of the strains, and determination of the bactericidal activity of the glutaraldehyde-based disinfectant used for automated endoscope reprocessing. In addition, patient outcome was analyzed by medical chart review, and incidence rates of clinical samples with P. aeruginosa were compared.
The University Hospital of Basel is an 855-bed tertiary care center in Basel, Switzerland. Approximately 1,700 flexible bronchoscopic, 2,500 gastroscopic, 1,400 colonoscopic, 140 endoscopic retrograde cholangiopancreatographic, and 140 endosonographic procedures are performed annually.
P. aeruginosa was detected in samples obtained from endoscopes in November 2009 for the first time since the initiation of surveillance in 2006. It was found in the rinsing water and in the drain of 1 of the 2 automated endoscope reprocessors. PFGE revealed 2 distinct P. aeruginosa strains, one in each reprocessor. The glutaraldehyde-based disinfectant showed no activity against the 2 pseudo-outbreak strains when used in the recommended concentration under standard conditions. After medical chart review, 6 patients with lower respiratory tract and bloodstream infections were identified as having a possible epidemiological link to the pseudo-outbreak strain.
This is the first description of a pseudo-outbreak caused by P. aeruginosa with reduced susceptibility to an aldehyde-based disinfectant routinely used in the automated processing of endoscopes.
Email your librarian or administrator to recommend adding this to your organisation's collection.