To send content items to your account,
please confirm that you agree to abide by our usage policies.
If this is the first time you use this feature, you will be asked to authorise Cambridge Core to connect with your account.
Find out more about sending content to .
To send content items to your Kindle, first ensure email@example.com
is added to your Approved Personal Document E-mail List under your Personal Document Settings
on the Manage Your Content and Devices page of your Amazon account. Then enter the ‘name’ part
of your Kindle email address below.
Find out more about sending to your Kindle.
Note you can select to send to either the @free.kindle.com or @kindle.com variations.
‘@free.kindle.com’ emails are free but can only be sent to your device when it is connected to wi-fi.
‘@kindle.com’ emails can be delivered even when you are not connected to wi-fi, but note that service fees apply.
In our study basic dosimetric properties of a flattening filter free 6 MV photon beam shaped by multileaf collimators (MLC) is examined using the Monte Carlo (MC) method.
Methods and Materials
BEAMnrc code was used to make a MC simulation model for 6 MV photon beam based on Varian Clinic 600 unique performance linac, operated with and without a flattening filter in beam line. Dosimetric features including central axis depth dose, beam profiles, photon and electron spectra were calculated and compared for flattened and unflattened cases.
Dosimetric field size and penumbra were found to be smaller for unflattened beam, and the decrease in field size was less for MLC shaped in comparison with jaw-shaped unflattened beam. Increase in dose rate of >2·4 times was observed for unflattened beam indicating a shorter beam delivery time for treatment. MLC leakage was found to decrease significantly when the flattening filter was removed from the beam line. The total scatter factor showed slower deviation with field sizes for unflattened beam indicating a reduced head scatter.
Our study demonstrated that improved accelerator characteristics can be achieved by removing flattening filter from beam line.
Email your librarian or administrator to recommend adding this to your organisation's collection.